Regulate preparation and Notice
1. Connection (please refer to the product mix content and connection method above).
2. Regulate the voltage selection, 115V for the standard voltage range for the 90-120V, 230V for 210-240V range. (This is very important, a wrong choice is easy to damage or affect the effectiveness of the instrument)
3. Confirm the main power switch off position at 0.
4. Plugin the power and turn the main power switch.
5. Gently coated with special rubber on the treating parts. (Cleaning before conditioning, better)
6. Installed the RF head on the probe handle. (Connect the metal cylinder into the connection hole to the deepest)
7. Subject's hand with a non-drying (dry cleaning) grip loop handle, thumb put on the red power button on the safe preparation, if not press the red switch, the machine will not work.
8. Regulate the time required.
Voltage & current: 90-120V/210-240V
Supreme power consumption: 350W
Function frequency: 3Mhz
Material: ABS resin (host)
NW: about 3.5KG
Packing specification: 350MM*350MM*280MM
GW: about 5KG